Portable Oxygen Concentrator
Portable Oxygen Concentrator
This Portable Oxygen Concentrator is light weight of 5.2 kg produces oxygen purity range at 90%+5%/-5% with the adjustable flow from 1 to 5 settings. It can be used by home power, car battery and lithium battery. If you have any other questions, Medicalbulkbuy is happy to answer them. Make sure reading and understanding all the information contained in this guide before you operate your unit. This was developed by medicalbulkbuy to meet the regular demand by customers for car operated,battery operated, light weight and carrying cart type at very reasonable price first time in India.
Difference between regular and this portable:
Persons using this can go with this machine to their office,travel in car, use in trains where ever you have power supply as the machine is having carrying cart like Suitcase and bag. Persons using will feel comfort and lead their life like anyone around them.
|Flow Settings||1 – 5 settings equal to 1-5LPM flow|
|Oxygen Purity||90%+5%/-5% at any settings|
|Outlet Pressure||8.5 PSI|
|Sound Level||≤ 40 dB (nominal) at flat table, ≤45dB when travelling|
|Size||9.1 in x 8.7 in x 12.8 in (23 cm x 22 cm x 32.5 cm)|
|Weight||11.5 lbs (5.2kg) Machine only, 13.7lbs(6.2kg) with battery|
|Remote controller||Wireless, 10 meters circle through the walls|
|User Interface||Push buttons, back-lit liquid crystal display (LCD)|
|Work Temperature||41° to 104°F (5° to 40°C)|
|Relative Humidity||15 to 95%|
|Altitude||Up to 12,000 ft (3657m)|
|Storage Temperature||-4° to 140°F (-20° to 60°C) unit only|
|Storage Humidity||Up to 95%, non condensing|
|AC Power Input||100 to 240 VAC, 50/60 Hz, 1.8 A max|
|AC Power Output||14.8 VDC, 8 A max|
|DC Power Input||Input 11. to 16.8 VDC 20 A max|
|DC Power Output||14.8 VDC, 8 A max|
|Chemistry of Battery||Lithium Ion 14.8 VDC (nominal)|
|Battery Dimensions||6.1 in x 3.4 in x 2 in(15.5 cm x 8.5 cm x 5 cm)|
|Battery Weight||1.5 lbs (0.7 kg)|
|Battery Charge Time||Approximately 4 to 5 hours|
|Battery Duration||Approximately Two hours at 1-5 any settings, extra battery is optional|
|O 2 concentration
|Green Light=Normal, Yellow=Caution <82%,
Red with Audible Alarm=Warning <75%
|Audible Alarm Sound||55 dB (nominal) at 39 in (1m) distance|
|Alarm type||Loss of power, High temperature, Low Therapeutic O 2 Out|
Important Safety Rules
When using electrical products, especially for children, basic safety precautions should always be followed. Read all instructions before using. Important information is highlighted by these terms:
DANGER To reduce the risk of serious or fatal injury from electrical shock: 1. Do not use while bathing. 2. Do not place the product where can be splashed liquid or fall into bathtub, water or other liquid. 3. In the event of unit falling into water, do not reach for the machine. Turn off the home power immediately then send it to dealer or our authorized service center for examination and repair.
WARNING 1. Check the electric rating in the unit label and wall outlet before connecting power cord. 2. Keep all electrical cords away from heated surfaces. 3. Do not operate this unit if it:
● has been dropped or knocked heavily.
● has been splashed liquid.
● has a damaged cord or plug.
● is not working properly.
Usage of the machine:
Portable Oxygen Concentrator is used between 1 ltr to 5 ltr. At 1 to 3 ltrs the green light is attained early and generally it takes 20 minutes time to get the green light at 4 and 5 ltr . So the yellow light glows and then after running the machine sufficient time, it gets the green light. So the attendant has to give oxygen to patient after getting the green light. Medicalbulkbuy checks the quality of the machine before dispatching the product as a final check on quality.
WARNING This device produces high levels of oxygen, which promotes rapid burning. Do not allow smoking or open flames within 5 feet of this device and any oxygen delivery accessory. Do not use oil, grease or petroleum-based products on or near the unit. Disconnect the power cord from the electrical outlet before you clean or repair the unit